Laser optical flow measurement
The laser diode pulsing system (LD-PS) by OPTOLUTION is a low-cost illumination for Particle Image Velocimetry (PIV). It generates bright and very thin light sheets with user selectable pulse spacing (up to 35 kHz) and pulse length. The system is very compact and extremely easy to use. The high brightness is achieved by toggling high-power OSRAM laser diodes using a special laser driver circuit. We offer optical power outputs of 5 W, 20 W and 40 W. The systems can be used for measurements in air or water. The LD-PS includes a built-in synchronizer that can be used to generate pulses for cameras. We have developed firmware for synchronizing the pco panda and the pco pixelfly and the OPTOcam 2/80 cameras in PIVlab. Additionally, the LD-PS can control the timing of the low-cost high-speed Chronos camera (Kron Technologies). But the laser can also be setup to accept TTL pulses from a third-party synchronizer. In this case, you can e.g. use the Q-Switch outputs from your existing setup to synchronize the LD-PS. Light sheet optics, an optical rail and a tripod mount (Manfrotto 200PL- Quick Release Plate) are incuded.
The settings of the LD-PS are changed wirelessly (USB radio dongle included) - directly in PIVlab, or in a custom software. The very compact LD-PS is actively cooled by two fans and has a very small average power consumption.
The most important advantage over continuous wave lasers is that frame straddling can be enabled, allowing to capture higher velocity events using very low interframe time (down to 10 µs interframe time is possible). The LD-PS is also order of magnitudes brighter than many CW lasers.
Aspects you have to keep in mind with pulsed laser diodes:
With pulsed lasers, significantly higher flow velocities can be measured than with CW lasers - also in combination with low frame rate cameras. This is enabled by using a technique called "frame straddling". For example, a speed of up to 20 m/s can be measured with the 5 Hz, 25-megapixel pco.panda. With the 1000 Hz, 1.2-megapixel Chronos 1.4, CW lasers would enable a maximum speed of 2 m/s. With a pulsed laser, speeds of up to 40 m/s are possible under the same conditions.
Brightness is determined by the pulse length. If you lower the pulse length, you will have to decrease the illuminated area to still have enough light.
Manual LD-PS/20 and 40 without synchronizer
Dr. William Thielicke